Frequency vs flux experiments#
In this section we show how to run experiments to study the flux dependence.
Flux-tunable transmons#
It can be shown for flux-tunable transmons that applying an external flux \(\Phi\) induces to a detuning on the transmon frequency [8]. By applying short flux pulses it is possible to tune the transmon frequency in order of GHz, which leads to several applications including quantum logical gates.
The transmon frequency as a function of the external flux can be expressed as [1]
where \(f_{\text{max}} = ( \sqrt{8 E_C E_J} - E_C) / h\) is the maximum qubit frequency, \(d\) is the junctions asymmetry, \(E_C\) is the charging energy, \(E_J\) is the Josephson energy and \(\Phi_0 = h / (2e)\) is the flux quanta.
The resonator detuning in the transmon regime \(E_J \gg E_C\) can be computed as
where \(f_r^{\text{bare}}\) is the bare frequency of the resonator and \(g_0^2\) is the coupling between the transmon and the resonator.
In Qibocal we provide two experiments to measure and fit the curves described by the two equations above. To measure the qubit detuning a 2D sweep is performed probing the systems at different drive frequencies and at different flux (bias) offset values. For the resonator we perform the same experiment by sweeping the readout frequency instead of the drive frequency.
Qubit flux dependence#
Parameters#
- class qibocal.protocols.flux_dependence.qubit_flux_dependence.QubitFluxParameters(freq_width: int, freq_step: int, bias_width: Optional[float] = None, bias_step: Optional[float] = None, drive_amplitude: Optional[float] = None, transition: Optional[str] = '01', drive_duration: int = 2000)[source]
QubitFlux runcard inputs.
- drive_amplitude: Optional[float] = None
Drive amplitude (optional). If defined, same amplitude will be used in all qubits. Otherwise the default amplitude defined on the platform runcard will be used
- transition: Optional[str] = '01'
Flux spectroscopy transition type (“01” or “02”). Default value is 01
- drive_duration: int = 2000
Duration of the drive pulse.
- hardware_average: bool = False
By default hardware average will be performed.
- freq_width: int
Width for frequency sweep relative to the readout frequency [Hz].
- freq_step: int
Frequency step for sweep [Hz].
- nshots: int
Number of executions on hardware
- relaxation_time: float
Wait time for the qubit to decohere back to the gnd state
Example#
A possible runcard to assess how the qubit frequency changes by varying flux is the following:
- id: qubit flux dependence
operation: qubit_flux
parameters:
bias_step: 0.001
bias_width: 0.05
drive_amplitude: 0.002
drive_duration: 4000
freq_step: 200000
freq_width: 10000000
nshots: 1024
relaxation_time: 20000
The expected output is the following:
From the acquired data this protocol estimates the flux insensitive point “sweetspot”, which corresponds to the flux value where the frequency is maximed, as well as the drive frequency and the diagonal crosstalk coefficient \(V_{ii}\). Both the sweetspot and the \(C_{ii}\) can be understood by writing the full expression for the flux felt by qubit \(i\) [1]:
where \(C_{ij}\) is known in the literature as the crosstalk matrix, while \(V_{i}\) is the applied voltage.
Requirements#
Resonator flux dependence#
Parameters#
- class qibocal.protocols.flux_dependence.resonator_flux_dependence.ResonatorFluxParameters(freq_width: int, freq_step: int, bias_width: Optional[float] = None, bias_step: Optional[float] = None)[source]
ResonatorFlux runcard inputs.
- freq_width: int
Width for frequency sweep relative to the readout frequency [Hz].
- freq_step: int
Frequency step for sweep [Hz].
- hardware_average: bool = False
By default hardware average will be performed.
- nshots: int
Number of executions on hardware
- relaxation_time: float
Wait time for the qubit to decohere back to the gnd state
Example#
A possible runcard to assess how the resonator frequency changes by varying flux is the following:
- id: resonator flux dependence
operation: resonator_flux
parameters:
bias_step: 0.01
bias_width: 0.4
freq_step: 100000
freq_width: 6000000
nshots: 2000
relaxation_time: 1000
From this protocol it is possible to extract both the bare and the dressed resonator frequency as well as an estimate for the coupling \(g_0\). It is suggested to run this protocol only after executing the qubit flux dependence experiment since some of the coefficients required can be computed with that experiment.
Requirements#
Qubit flux dependence (tracking)#
As we saw above both the resonator and the qubit are affected by an external flux. In the qubit flux dependence the measurement is performed at fixed readout frequency. To take into account how the transmon shifts in frequency when probing the qubit we have another experiment qubit_flux_tracking.
Parameters#
- class qibocal.protocols.flux_dependence.qubit_flux_tracking.QubitFluxTrackParameters(freq_width: int, freq_step: int, bias_width: Optional[float] = None, bias_step: Optional[float] = None, drive_amplitude: Optional[float] = None, transition: Optional[str] = '01', drive_duration: int = 2000)[source]
QubitFluxTrack runcard inputs.
- bias_step: Optional[float] = None
Bias step for sweep [a.u.].
- bias_width: Optional[float] = None
Width for bias sweep [V].
- drive_amplitude: Optional[float] = None
Drive amplitude (optional). If defined, same amplitude will be used in all qubits. Otherwise the default amplitude defined on the platform runcard will be used
- drive_duration: int = 2000
Duration of the drive pulse.
- hardware_average: bool = False
By default hardware average will be performed.
- transition: Optional[str] = '01'
Flux spectroscopy transition type (“01” or “02”). Default value is 01
- freq_width: int
Width for frequency sweep relative to the readout frequency [Hz].
- freq_step: int
Frequency step for sweep [Hz].
- nshots: int
Number of executions on hardware
- relaxation_time: float
Wait time for the qubit to decohere back to the gnd state
Example#
Here is a possible runcard:
- id: qubit flux dependence
operation: qubit_flux_tracking
parameters:
bias_step: 0.001
bias_width: 0.05
drive_amplitude: 0.002
drive_duration: 4000
freq_step: 200000
freq_width: 10000000
nshots: 1024
relaxation_time: 20000